New Firmware is available from Instantel for both the Micromate and Minimate Pro units.
KANATA, Ont. – Instantel has released new firmware for both its Micromate and Minimate Pro monitoring units.
MM10.88 for Micromate units corrects two issues. First, that of the histogram peak of the sound level microphone was higher than the waveform peak, when recording in histogram combo mode. This occurred when the peak was recorded at the end of the interval. Second, the corruption of histogram event files. This occurred when the Micromate was rebooted while it was recording in histogram mode.
The release of version FW10.75 for all models of Minimate Pro monitoring units corrects an issue where auto call home feature would not work when the auto call home access type was set to DNS name on the Minimate Pro monitor. The Minimate Pro was not correctly converting the call home DNS name to the corresponding IP address. Auto call home access types set to IP address were not affected.
Micromate and Minimate Pro users can get a copy of the latest firmware at www.Instantel.com/resources and click on firmware. From the list of available firmwares, click on the download link beside either the Micromate or the Minimate Pro firmware.
For instructions on upgrading your monitoring unit, please consult the release notes. Download the Micromate firmware release notes here. Download the Minimate Pro firmware release notes here.